Manuale Medisana BU 510. Visualizza gratuitamente il manuale Medisana BU 510 oppure richiedilo ad altri proprietari Medisana BU 510. Maximum error tolerance for static pressure: ± 3 mmHg. Maximum error tolerance for pulse rate: ± 5 % of the value. Measuring method: Oscillometric. ERR 3 If the cuff takes too long to inflate, it is possible that the cuff is incorrectly. positioned or that the connection with the hose is not airtight.
Doubt. pity: Medisana error 3
|ERROR LOADING FEED DATA JOOMLA|
|SQL ERROR 9260|
|Errors were encountered while processing mysql-server-5.5|
|Bcbsmp.lib link error|
youtube videoError 3 - electronic blood pressure monitor
Blood pressure monitor
Please read carefully!
Explanation of symbols
Follow the instructions for use!
Non-observance of these instructions can
result in serious injury or damage
to the device.
These warning notes must be observed to
prevent any injury to the user.
These notes must be observed to prevent
any damage to the device.
These notes give you useful additional
information on the installation or
Device classification: type BF applied part
Date of manufacture
48686 (51172)05/2018 Ver. 1.0
• The device is intended for measuring the blood
pressure from the upper arm of an adult.
• The device is not suitable for measuring the
blood pressure of children. Ask your doctor
before using it on older children.
• Persons who suffer from arrhythmia, diabetes,
cardiovascular problems or who have had a stroke
should consult their doctor before using the device.
Read the instruction manual carefully before
using this device, especially the safety
instructions, and keep medisana error 3 instruction manual
for future use. Should you give this device to
another person, it is vital that you also pass on
these instructions for use.
• The unit is intended only for use in the home. Consult your doctor prior to
using the blood pressure monitor if you are concerned about health matters.
• Only use the unit for its intended purpose in accordance with the instructions
• Warranty claims become void if the unit is misused, medisana error 3.
• Please consult your doctor prior to using the unit if you suffer from illnesses
such as arterial occlusive disease.
• The unit may not be used to check the heart rate of a pacemaker.
• Pregnant women should take the necessary precautions and pay attention to
their individual circumstances. Please consult your doctor if necessary.
• If you experience discomfort during a measurement, such as pain in the
upper arm or other complaints, press the START button
to release the air
immediately from the cuff. Loosen the cuff and remove it from your arm.
• Use the right cuff, otherwise it can not work.
• The unit is unsuitable for children.
• Do not allow children to use the unit. Medical products are not toys!
• Please ensure that the unit is kept away from the reach of children, medisana error 3.
• The cuff hose around the neck may cause suffocation.
• The swallowing of small part like packaging bag, battery, medisana error 3, battery cover and so
on may cause the suffocation.
• The user must check that the equipment functions safely and see that it is in
proper working condition before being used.
• Only the cuff provided can be used, medisana error 3, and cannot be changed to any other
branded cuff. It can only be replaced with a cuff of exactly the same type.
• The device may not be operated in rooms with high levels of radiation or in
the vicinity of high-radiation devices, medisana error 3 as radio transmitters, mobile
telephones or microwaves, as this could cause functional defects or incorrect
• Do not use the equipment where flammable gas (such as anaesthetic gas,
oxygen or hydrogen) or flammable liquid (such as alcohol) are present.
• No modification of this unit/device is allowed.
• Please do not attempt to medisana error 3 the unit yourself in the event of malfunctions, medisana error 3,
since this will void all warranty claims. Only have repairs carried out by
authorised service centres.
• Protect the unit against moisture. If liquid does manage to penetrate into the
unit, remove the batteries immediately and do not continue to use the unit. In
this case contact your specialist dealer or notify us directly.
• Please don’t use a dilution agent, alcohol or petrol to clean the unit.
• Please don’t hit heavily or fall down the product from a high place.
• If the unit is not going to be used for a long period, please remove medisana error 3
SAFETY NOTES FOR Medisana error 3
• Do not disassemble batteries!
• Replace the new batteries if the unit display a low battery symbol.
• Never leave any low battery in the battery compartment since they may leak
and cause damage to the unit.
• Increased risk of leakage, avoid contact with skin, eyes and mucous
membranes! If battery acid comes in contact with any of these parts, rinse
the affected area with copious amounts of fresh water and seek medical
• If a battery has been swallowed seek medical attention immediately!
• Replace all of the batteries simultaneously!
• Only replace with batteries of the same type, never use different types of
batteries together or used batteries with new ones!
• Insert the batteries correctly, observing the polarity!
• Please take off the medisana error 3 if you won’t use in 3 months.
• Remove the batteries from the device if it is not going to be used for an
• Keep batteries out of children›s reach!
• Do not attempt to recharge these batteries! There is a danger of explosion!
• Do not short circuit! There is a danger of explosion!
• Do not throw into a fire! There is a danger of explosion!
• Do not throw used batteries into the household refuse; put them in a
hazardous waste container or take them to a battery collection point, at the
shop where they were purchased!
Unit and LCD display
Socket for mains adaptor
Blood Pressure Monitor
Battery Compartment (on
Push-in Connector for air tube
LCD screen (display)
Display of Date/Time
Display of Systolic
Display of Diastolic Pressure
Display of Pulse
Change Battery symbol
Arrhythmia display (arrhythmia cordis)
Memory Location number
Blood Pressure Indicator
(green - yellow - orange - red)
Scope of supply
Please check first of all that the unit is complete.
The following parts are included as standard:
• 1 MEDISANA blood pressure monitor BU A50
• 1 cuff with air tube
• 4 batteries (type AA, LR6) 1.5 V
• 1 storage pouch
• 1 instruction manual
Please contact your supplier if you notice any transport damage on unpacking
What is blood pressure?
Blood pressure is the pressure created in the blood vessels by each heart beat.
When the heart contracts (= systole) and pumps blood into the arteries, it brings
about a rise in pressure. The highest value is known as the systolic pressure and
is the first value to be measured. When the cardiac muscle relaxes to take in fresh
blood, the pressure in the arteries also falls. Once the arteries are relaxed, a
second reading is taken – the diastolic pressure.
6. The readings are automatically saved in the selected memory (, or
). Up to 60 measured values with date and time can be stored in each memory.
When the memory is full, the oldest reading is always the rst to be deleted.
7.The measurement readings remain on the display. The unit switches o
automatically after approx. 3 minutes if no buttons are pressed. The unit may medisana error 3 be switched o using the START/STOP button
Discontinuing a measurement
If it is necessary to interrupt a blood pressure measurement for any reason
(eg. the patient feels unwell) the START/STOP button
can be pressed at any
time. The device medisana error 3 decrease the cu pressure automatically.
Displays of stored values
This unit features 4 separate memories, each with a capacity of 60 memory slots.
Results are automatically stored in the memory selected by the user. Press
the + or - button when power off, the latest 3 times average will be shown,
press the + button again, the oldest measurement value will be shown, or press
the - button again, the latest measurement will be shown, as well as subsequent
measurements can be displayed one after the other by pressing the + or -
button each time. If you have reached the final entry and do not press a button,
the unit automatically switches itself off after approximately 120 seconds in
memory recall mode, medisana error 3. Memory recall mode can be exited at any time by pressing
the START/STOP button
, which will also switch the unit off. The oldest value
is deleted if 60 measured values have been stored in the memory and a new value
Memory - clear of measurements
If you are sure that you want to permanently remove all stored memories, press
the SET button
for medisana error 3 times until CL appears. Press the START/STOP button
CL will flash for 3 times to clear all the memories. After this press +- oder -
button, Mand ″no″ will be shown on the display which mean that no memory
Error messages and error remedying
The following symbol will appear on the display when measuring abnormal.
Symbol Cause Correction
E-1Weak signal or Wrap the cuff properly.
pressure change Remeasure with correct way.
E-2External strong When near cell phone or other high radiant
disturbance device, medisana error 3, the measurement will be failed.
Keep quiet and no chatting when measure.
E-3It appears error Wrap the cuff properly.
during the process Make sure that he plug is properly inserted
of inflating in the unit.
E-5Abnormal blood Repeat the measurement after relax for
pressure 30 mins. If get unusual readings for 3 times,
please contact your doctor.
Low battery The batteries are too low or empty.
Replace all four batteries with new 1.5 V
type AA LR6 batteries.
Problem Check Cause and solutions
No power Check the battery power Replace new one
Check the polarity position Installation for proper placement
of the batteries polarities
No inflation Check whether the plug is Insert into the air socket tightly
Check whether the plug is Change a new cuff
broken or leak
Err and stop Check whether move the arm Keep the body peaceful
working when inflate
Check if chatting when Keep quiet when measure
Cuff leak Check whether the cuff Wrap the cuff tightly
wrap too loose
Check whether the cuff Change a new cuff
Please contact the distributer if you can’t solve the problem. Do not disassemble
the unit by yourself.
Cleaning and care
Remove the batteries before cleaning the unit. Clean the unit using a soft
cloth lightly moistened with a mild soapy solution. Never use abrasive cleaning
agents, alcohol, naphta, thinner or gasoline etc. Never immerse the unit or any
component in water. Be cautiious not to get any moisture in the main unit.
Do not wet the cuff or attempt to clean the cuff with water. Using a dry cloth,
gently wipe away any excess moisture that may remain on the cuff. Lay the cuff
flat in an unrolled position and allow the cuff to air dry.
Do not expose the unit to direct sunlight; protect it against dirt and canon mp190 e13 error.
Do not subject the unit to extreme hot or cold temperature. Keep the unit in the
storage case when no use. Store the unit in a clean and dry location.
This product must not be disposed of together with domestic waste.
All users are obliged to hand in all electrical or electronic devices,
regardless of whether or not they contain toxic substances, at a
municipal or commercial collection point so that they can be disposed
of in an environmentally acceptable manner.
Remove the batteries before disposing of the device/unit. Do not dispose of
old batteries with your household waste, but at a battery collection station at a
recycling site or in a shop.
Consult your local authority or your supplier for information about disposal.
Guidelines / Standards
This blood pressure monitor meets the requirements of the EU standard for non-
invasive blood pressure monitors. It is certified in accordance with EC Guidelines
and carries the CE symbol (conformity symbol) “CE 0297”.
The blood pressure monitor corresponds to European standards EN 1060-1 and
EN 1060-3. The specifications of EU Guideline “93/42/EEC of the Council
Directive dated 14 June 1993 concerning medical devices” are met and likewise
those of EU Guideline 89/336/EEC.
Electromagnetic compatibility: (see attached leaflet)
Name and model : MEDISANA Blood pressure monitor BU A50
Display system : Digital display
Memory slots : 4 x 60 measurement values
Measuring method : Oscillometric
Voltage supply : 6 V=, 4 x 1.5V battery AA LR6
Measuring range blood
pressure : 0 – 299 mmHg
Measuring range - pulse : 40 – 199 beats/min.
deviation of static pressure : ± 3 mmHg
deviation of pulse values : ± 5 % of the value
Pressure generation : Automatic with pump
Air release : Automatic
Autom, medisana error 3. switch-off : After 3 min.
Operating conditions : +5 °C to +40 °C, medisana error 3,
15 to 85 % max. relative humidity
Storage conditions : -20 °C to +55 °C,
10 to 85 % max. relative humidity
Dimensions (L x W medisana error 3 H) : approx. 139 x 100 x 46 mm
Cuff : 500 x 150 mm / 22 – 36 cm for adults
Weight (main unit) : approx. 292 g without batteries
Item number : 48686
EAN number : 40 15588 48686 9
Accessories : - Mains adapter
Art. No. 51125 / EAN 40 15588 51125 7
- Cuff M 22 – 36 cm for adults with
average upper arm circumference
Art. No. 51168 / EAN 40 15588 51168 4
- Cuff, L 32 - 42 cisco #send errors 10, #recv errors 0 for adults with upper
arm of large circumference
Art. No. 51169 / EAN 40 15588 51169 1
In accordance with our policy of continual product improvement, we
reserve the right to make technical and visual changes without notice.
The current version of this instruction manual can be found under medisana error 3 and repair terms
Please contact your supplier or the service centre in case of a claim under the
warranty. If you have to return the unit, please enclose a copy of your receipt and
state what the defect is.
The following warranty terms apply:
1. The warranty period for MEDISANA products is three years from date of
purchase. In case of a warranty claim, the date of purchase has to be proven
by means of the sales receipt or invoice.
2. Defects in material or workmanship will be removed free of charge within the
3. Repairs under warranty do not extend the warranty period either for the unit
or for the replacement parts.
4. The following is excluded under the warranty:
a. All damage which has arisen due to improper treatment, e.g. nonobservance
of the user instructions.
b.All damage which is due to repairs or tampering by the customer or
unauthorised third parties.
c.Damage which has arisen during transport from the manufacturer to the
consumer or during transport to the service centre.
d.Accessories which are subject to normal wear and tear as batteries, cu etc.
5.Liability for direct or indirect consequential losses caused by the unit are
excluded even if the damage to the unit is accepted as a warranty claim.
MEDISANA GmbH • Jagenbergstraße 1 9 • 41468 NEUSS • GERMANY
Common factors medisana error 3 wrong measurement
• Please keep quiet for 5-10 minutes and avoid
eating, drinking alcohol, smoking, medisana error 3, exercising
and bathing before taking measurement. All
these factors will influence the measurement
• Remove any garment that fits closely to your
• Always measure on the same arm (normally left).
• Take measurement regularly at the same time of
every day, medisana error 3, as blood pressure changes even during
• All efforts by the patient to support their arm can
increase blood pressure.
• Make sure medisana error 3 are in a comfortable, relax position
and do not activate any of the muscles in the
measurement arm during measurement. Use a
cushion for support if necessary.
• If the arm artery lies lower or higher than the
heart, a false reading will be obtained.
• A loose or open cuff causes false reading.
• With repeated measurementsblood accumulates
in the arm which can lead to false reading.
Consecutive blood pressure measurements should
be repeated after 1 minute pause or medisana error 3 the
arm has been held up in order to allow the
accumulated blood to flow away.
Influencing and evaluating readings
• Measure your blood pressure several times, then record and compare the
results. Do not draw any conclusions from a single reading.
• Your blood pressure readings should always be evaluated by a doctor who is
also familiar with your personal medical history. When using the unit regularly
and recording medisana error 3 values for your doctor, you should visit the doctor from time
to time to keep him updated.
• When taking readings, remember that the daily values are influenced by
several factors. Smoking, consumption of alcohol, drugs and physical exertion
influence the measured values in various ways.
• Measure your blood pressure before medisana error 3.
• Before taking readings, allow yourself at least 5-10 minutes rest.
• If the systolic and diastolic readings seem unusual (too high or too low) on
several occasions, despite correct use of the unit, please inform your doctor.
This also applies to the rare occasions when an irregular or very weak pulse
prevents you from taking readings.
Insert / replace batteries
You must insert the batteries provided before you can use your unit. Medisana error 3 lid medisana error 3
the battery compartment
is located on the underside of the unit. Open it,
remove it and insert the 4 AA LR6 type 1.5 V batteries supplied. Ensure correct
polarity when inserting the batteries (as marked inside the battery compartment).
Close the battery compartment. Replace the batteries if the change battery
appears on the display
or if nothing appears on the display after
the unit medisana error 3 been switched on.
Using a Mains Adaptor
Alternatively, medisana error 3, you can use the unit with a special mains adaptor (MEDISANA
Art.–No. 51125). This connects to the socket
provided at the back of the
unit. In this case the batteries remain in the unit. Inserting the connector into
the back of the blood pressure monitor switches the batteries off mechanically.
It is necessary, medisana error 3, therefore, to plug the mains adaptor into the mains socket
first of all and then to connect it to the blood pressure monitor. If the blood
pressure monitor is no longer being used, first remove the connector from
the blood pressure monitor and then pull the mains adaptor out of the mains
socket. Doing this means that the date and time will not have to be reset each
1. User setting:
Press SET button
when power off. Continue to above step, the screen will
displaypress + or - button, it will be changed between user and user
and and. Press SET button
when you confirm the user, then it will enter
into the year setting mode.
NOTE: press the button
, to quick change the user when power off.
2, medisana error 3. Year setting:
The input for setting the year then starts flashing. Keep pressing the + or -
button until the required year appears. Press SET button
when you confirm
the year, then it will switch to the month and date setting mode.
3. Month and date setting:
The input medisana error 3 setting the month then starts flashing. Keep pressing the + or -
button until the required month appears. Press SET button
when you confirm
the month, then it will set the date.
As with the month setting, keep pressing the + or - button until the required
day appears. Press SET button
when you confirm the date, then it will switch
to motorola critil error time setting mode.
4. Time setting:
The input for setting the hour then starts medisana error 3. Keep pressing the + or - button
until the required hour appears. Press SET button
when you confirm the hour.
The input for setting the minute then starts flashing. As with the hour setting,
keep pressing the + or - button until the required minute appears. Press SET
when you confirm the minute, which will complete the setting mode.
CL appears in the display. You can quit set-up mode by pressing SET button
If the batteries are changed the medisana error 3 is lost and the date and time must
be set again.
Fitting the cuff
1.Push the end piece of the air tube
into the hole on the left side of the unit
prior to use.
2. Slide the open end of the cuff through the metal bracket so that the Velcro
fastener is on the outside and it becomes a cylindrical form (Fig.1). Slide the
cuff over your left upper arm.
3.Position the air hose in the middle of your arm in line with your middle finger
(Fig.2) (a). The lower edge of the cuff should be 2 - 3 cm above the crease of
the elbow. (b). Pull the cuff tight and close the Velcro fastener. (c).
4.Measure the pulse on your bare arm.
5. Only position the cuff on the right arm if it cannot be used on the left arm.
Always carry out measurements on the same arm, medisana error 3.
6.Correct measuring position for sitting (Fig.3).
Blood pressure that is too low represents just as great a health
risk as blood pressure that is too high! Fits of dizziness may lead
to dangerous situations arising (e.g, medisana error 3. on stairs or in traffic)!
How is blood pressure measured?
The MEDISANABU A50 is a blood pressure unit which is used to measure blood
pressure at the upper arm. The measurement is carried out by a microprocessor,
which, via a pressure sensor, measures the vibrations resulting from the inflation
and deflation of the cuff over the artery.
Blood pressure classification WHO
These values have been specified by the World Health Organisation (WHO)
without making any allowance for age.
Low blood medisana error 3 <100 diastolic <60
Normal blood pressure(green display area
systolic 100 - 139 diastolic 60 - 89
Forms of high blood pressure / hypertension
Mild hypertension(yellow display area
systolic 140 – 159 diastolic 90 – 99
Moderate hypertension(orange display area
systolic 160 – 179 diastolic 100 – 109
Severe hypertension(red display area
systolic >_180 diastolic >_110
Do not take any therapeutic measures on the basis of a self
measurement. Never alter the dose of a medicine prescribed
by a doctor.
Taking a blood pressure measurement
After the cuff has been appropriately positioned,the measurement can begin:
1.Switch the unit on by pressing the START/STOP button
2, medisana error 3. If the START/STOP button
is pressed, two short beeps are heard and all
display characters are shown (display test). This test can be used to check that
the display is indicating properly and in full.
3.The unit is ready for measurement and the number 0 flashes for 2 seconds. It
automatically inflates the cuff slowly in order to measure your blood pressure.
The rising pressure in the cuff is shown on the display.
4.The unit inflates the cuff until sufficient pressure has built up for a
measurement. Then the unit slowly releases air from the cuff and carries
out the measurement. When the device detects medisana error 3 signal, the heart symbol
on the display starts to flash, you can hear the beep for every
heartbeat once the heartbeat signal is detected.
5.When the measurement is finished, you can hear a long beep and the cuff is
deflated completely. The systolic and diastolic blood pressure and the pulse
value with the pulse symbol appear on the display
. The blood pressure
flashes next to the relevant coloured bar depending on the WHO
blood pressure classification. If the unit has detected an irregular heartbeat,
the arrhythmia display also flashes.
6.1 Guide to display symbols
Error during measurement
Pulse is lower than 40 beats/minute
Batteries low. Please replace batteries.
If the display shows one of the following error messages, this means that the unit
was unable to accurately determine blood pressure.
Possible causes and solutions
ERR 1 Measurement of the systolic blood pressure was completed.
Either the connection with the hose has been interrupted or no pulse could
Check the connections between the cuff and the main unit.
ERR 2 Non-physiological pressure has been detected.
Cause: the arm was moved during the measurement.
Repeat the measurement, keeping your arm still.
ERR 3 If the cuff takes too long to inflate, it is possible that the cuff is incorrectly
positioned or that the connection with the hose is not airtight.
Check the connections and repeat the measurement.
ERR 4 The difference between systolic and diastolic pressure is abnormally high.
Repeat the measurement under relaxed conditions, following the
If the same error occurs repeatedly with further readings, please contact
ERR 5 An average value could not be calculated due to unstable conditions while
the readings were being taken.
Repeat the measuring process, keeping your arm still and not speaking
Medisana MTC 51139 User Manual
Name and model
: MEDISANA blood pressure monitor MTC
: Digital display
: 6 V= 4 x 1,5 V Mignon AA LR 6, Alkaline
Blood pressure measuring range
: 0 – 300 mmHg
Pulse measuring range
: 30 – 180 beats / min.
Maximum error tolerance for static pressure
: ± 3 mmHg
Maximum error tolerance for pulse rate
: ± 5 % of the reading
: automatic, using pump
: after approx. 1 minute
: + 5 °C to + 40 °C,
< 85 % humidity
: – 20 °C to + 55 °C
< 95 % humidity
: approx. 108 mm x 151 mm x 63 mm
: 22 – 30 cm Adult cuff for average
upper arm circumference
: approx. 300 g without batteries
: 40 15588 51139 4
: Mains adapter
EAN-No. 40 15588 51125 7
Cuff, medisana error 3, large 30 - 42 cm for
adults with upper arm of large circumference
EAN-Nr. 40 15588 51126 4
In accordance with our policy of continual product improvement,
we reserve the right to make technical and optical changes
51139_MTC_West_aktual.qxd:Blutdruckmessgerät 07.12.2010 12:03 Uhr Seite 23
Medisana BU 540 White, Arm blood pressure monitor, Bluetooth
Code - 51182
Design > Colour of product - White
Display > Display - Extra large, easy-to-read digital display.
EAN - 4015588511820
Link - https://www.medisana.com/en/Health-control/Blood-pressure-monitor/BU-540-connect-Upper-arm-blood-pressure-monitor.html
Name - Medisana BU 540 White, Arm blood pressure monitor, Bluetooth
Other features > Dimensions (WxDxH) - 140 x 130 x 60 mm
Other features > Power supply type - Batteries AAA LR03
Package features > Embeeded battery - No
Package features > Gross depth (mm) - 133 mm
Package features > Gross depth master carton - 440 mm
Package features > Gross height (mm) - 160 mm
Package features > Gross height master carton - 362 mm
Package features > Gross weight - 0.656 kg
Package features > Gross width (mm) - 100 mm
Package features > Gross width master carton - 295 mm
Package features > Net weight master carton - 10.496 kg
Package features > Packing quantity - 16 pc(s)
Package features > Paper/Pasteboard - 48 g
Package features > Plastic (No PET) - 12 g
Package features > Tare weight (kg) - 0.06 kg
Package features > Tare weight master carton - 0.799 kg
Package features > Volume (m3) - 0.002128 m³
Package features > WEE classification - CL109:5:2017-04-01
Package features > WEEE tax - Yes
Power > Battery type medisana error 3
Power > Battery voltage - 1.5 V
Power > Number of batteries supported - 4
Producer product name - BU 540
Technical details > Functions - Bluetooth
Technical details > Net weight - 0.596 kg
Technical details > Number of users -
Technical details > Other features - Blood pressure measuring range: 0 – 299 mmHg. Maximum error tolerance for static pressure: ± 3 mmHg. Maximum error tolerance for pulse rate: ± 5 % of the value. Measuring method: Oscillometric. Pressure generation: Automatic with pump. Deflation: Automatic. 250 memory spaces for each of 2 users. Very large cuff (arm circumferences 22 – 42 cm).
Technical details > Placement supported - Upper arm
Technical details > Pulse rate measuring - 40 – 199 beats/min. Arrhythmia detection.
Technical details > Type - Arm blood pressure monitor
Warranty - 24
Warranty - 24 month(s)
Adding product to your cartSKU: A268686
Gross depth (mm): 160.00
Gross height (mm): 10.00
Gross weight: 0.70
Gross width (mm): 135.00
Net weight: 0.64
Packing quantity: 1.00
Tare weight (kg): 0.06
Volume (m3): 0.00
Embeeded battery: No
WEEE tax: Yes
Type: Arm blood pressure monitor
Producer product name: BU 514
Display: Digital display.
Pulse rate measuring: 40 – 199 beats/min. Arrhythmia detection.
Colour of product: White
Number of batteries supported: 4
Power supply type: Batteries AA LR6
Battery voltage: 1.5
Dimensions (WxDxH): 130 x 110 x 60
Other features: Blood pressure measuring range: 0 – 299 mmHg. Maximum error tolerance for static pressure: ± 3 mmHg. Maximum error tolerance for pulse rate: ± 5 % of the value. Measuring method: Oscillometric. Pressure generation: Automatic with pump. Deflation: Automatic. 90 memory slots for each of 2 users. Very large cuff (arm circumferences 22 – 36 cm).
Placement supported: Upper arm
Plastic (No PET): 12
WEE classification: CL109:5:2017-04-01
Warranty: 24 medisana error 3